Diagnostics Portfolio You are about to exit the Abbott family of websites for a 3rd party website Links which take you out of Abbott worldwide websites are not under the control of Abbott, and Abbott is not responsible for the contents of any such site or any further links from such site. Abbott is providing these links to you only as a convenience, and the inclusion of any link does not imply endorsement of the linked site by Abbott. The website that you have requested also may not be optimised for your screen size. Do you wish to continue and exit this website? Yes No You are about to enter an Abbott country or region specific website.
|Published (Last):||9 September 2015|
|PDF File Size:||9.2 Mb|
|ePub File Size:||12.72 Mb|
|Price:||Free* [*Free Regsitration Required]|
The system can be configured with up to different assays including Ion Selective Electrode ISE analysis of sodium, potassium, and chloride with up to 90 different cartridges onboard. Individual assay time is a maximum of Operational modes include random, continuous access and priority processing. Endpoint and rate reactions are calculated using absorbance measurements that are converted into concentration units.
Absorbance measurements are made using a diode array detection system. The Integrated Chip Technology ICT for electrolyte determinations consists of an integrated module using ion selective electrode technology. Quality control using Levy-Jennings and Westgard rules and on-board data storage is located on the hard drive of the SCC. The cuvettes are held in 9 cuvette holders segments designed for the c system.
Each of the 9 segments contain 11 cuvettes. Each individual cuvette is read as it passes through the optical light path. A water distribution system is used to wash the cuvettes, sample probe, reagent probes and mixer blades. The cuvette washer is used to distribute diluted alkaline and acid solutions during the cuvette wash process.
The mixer assembly utilizes 2 mixers to mix the sample and reagent in preparation for final analysis. The c has 1 reagent supply center that contains both the R1 and R2 reagents. The chassis and external cover are stripped and or sanded and painted. All tubing, with the exception of some Teflon tubing if found to be in excellent condition are replaced. Next the instrument will be re-assembled, inspected and put thru the same verification procedures as the instrument went through initially at the factory, including clinical background check, carryover testing, precision, calibration and three levels of control testing.
In the end, a certificate of quality assurance will be attached to the report of all test result and diagnostic testing and a refurbish record will be created for each instrument. Reagents, controls, and consumables may be available. Service contracts may be available.
Abbott Architect c4000
- FRUCTOSEFREIE LEBENSMITTEL LISTE PDF
- KURS FOTOSHOPA PDF
- DIAGRAMMA DELLA VERITA PDF
- DEFINITIVNI VODIC KROZ GOVOR TELA PDF
- KANT GROUNDWORK OF THE METAPHYSICS OF MORALS MARY GREGOR PDF
- ATARI ST INTERNALS PDF
- EJERCICIOS RESUELTOS DE FUNCIONES INYECTIVAS SOBREYECTIVAS Y BIYECTIVAS PDF
- LEGO 10220 INSTRUCTIONS PDF
- HETEROCOSMICA DOLEZEL PDF