All rights reserved. Printed in U. Attention Refer to the manual see the Index for references. This symbol indicates that information about usage of a feature is contained in the manual. This symbol appears on the rear panel ground terminal and by the fuse compartment. AC Power Source The instrument is intended to operate from an ac power source that will not apply more than V ac rms between the supply conductors or between either supply conductor and ground.
|Published (Last):||27 September 2014|
|PDF File Size:||15.33 Mb|
|ePub File Size:||16.37 Mb|
|Price:||Free* [*Free Regsitration Required]|
All rights reserved. Printed in U. All product names are trademarks of their respective companies. The warranty period is one year and begins on the date of shipment. Parts, product repairs and services are warranted for 90 days. Fluke warrants that software will operate substantially in accordance with its functional specifications for 90 days and that it has been properly recorded on non-defective media.
Fluke does not warrant that software will be error free or operate without interruption. Fluke authorized resellers shall extend this warranty on new and unused products to end-user customers only but have no authority to extend a greater or different warranty on behalf of Fluke. Warranty support is available if product is purchased through a Fluke authorized sales outlet or Buyer has paid the applicable international price.
To obtain warranty service, contact your nearest Fluke authorized service center or send the product, with a description of the difficulty, postage and insurance prepaid FOB Destination , to the nearest Fluke authorized service center.
Fluke assumes no risk for damage in transit. Following warranty repair, the product will be returned to Buyer, transportation prepaid FOB Destination. If Fluke determines that the failure was caused by misuse, alteration, accident or abnormal condition of operation or handling, Fluke will provide an estimate of repair costs and obtain authorization before commencing the work.
Following repair, the product will be returned to the Buyer transportation prepaid and the Buyer will be billed for the repair and return transportation charges FOB Shipping Point. Since some countries or states do not allow limitation of the term of an implied warranty, or exclusion or limitation of incidental or consequential damages, the limitations and exclusions of this warranty may not apply to every buyer.
If any provision of this Warranty is held invalid or unenforceable by a court of competent jurisdiction, such holding will not affect the validity or enforceability of any other provision. Fluke Corporation P. Box Everett, WA U. Service Information General Specifications DC Voltage Specifications DC Current Specifications Resistance Specifications AC Voltage Sinewave Specifications AC Current Sinewave Specifications Capacitance Specifications Temperatore Calibration Thermocouple Specifications DC Power Specification Summary Power and Dual Output Limit Specifications Calculating Power Uncertainty Additional Specifications Frequency Specifications Harmonics 2nd to 50th Specifications AC Voltage, Squarewave Characteristics AC Voltage, Trianglewave Characteristics typical AC Current, Squarewave Characteristics typical AC Current, Trianglewave Characteristics typical DDS Assembly A Current Assembly A Voltage Assembly A Power Supplies Outguard Supplies Inguard Supplies Equipment Required for Calibration and Verification Starting Calibration How the Calibration Procedure Works DC Volts AC Volts Thermocouple Measuring DC Current AC Current Capacitance, Four-Wire Comp Remote Commands for A Calibration Generating a Calibration Report Calibration Shifts Report, Printout Format Calibration Shifts Report, Spreadsheet Format Calibration Constant Report, Printout Format Calibration Constants Report, Spreadsheet Format Performance Verification Tests Zeroing the Calibrator DC Current Amplitude Accuracy Resistance Accuracy Resistance DC Offset Measurement AC Current Amplitude Accuracy Capacitance Accuracy Thermocouple Measurement Accuracy Thermocouple Sourcing Accuracy Thermocouple Measuring Accuracy Phase and Frequency Accuracy Access Procedures Removing Analog Modules Removing Rear Panel Assemblies Removing the Keyboard and Accessing the Output Block Diagnostic Testing Running Diagnostics Sequence of Diagnostics Tests Diagnostics Error Messages Testing the Front Panel Internal Fuse Replacement Complete List of Error Messages DC Voltage Offset Accuracy How to Obtain Parts How to Contact Fluke Parts Lists SC Specifications Volt Specifications Edge Specifications Leveled Sine Wave Specifications Time Marker Specifications Wave Generator Specifications Pulse Generator Specifications Trigger Signal Specifications Pulse Function Trigger Signal Specifications Edge Function Trigger Signal Specifications Oscilloscope Input Resistance Measurement Specifications Oscilloscope Input Capacitance Measurement Specifications Overload Measurement Specifications Theory of Operation Voltage Mode
Fluke 5500A Getting Started
FLUKE 5500A Operator's Manual