Boulder, CO: Medtronic April 8 — May 6, October and February Based on internal test report RE, LF design verification report: benchtop testing using porcine abdominal aorta, mesentery, and renal arteries, with average lbs.
|Published (Last):||3 February 2018|
|PDF File Size:||3.24 Mb|
|ePub File Size:||12.81 Mb|
|Price:||Free* [*Free Regsitration Required]|
Boulder, CO: Medtronic April 8 — May 6, October and February Based on internal test report RE, LF design verification report: benchtop testing using porcine abdominal aorta, mesentery, and renal arteries, with average lbs. Instances of tissue sticking to jaws measured over seals per device. October 5, Based on internal test report RE, Bag and device reliability memo. April 13, April 10 to 12, Based on internal report RE, Lig product claim LF benchtop testing using porcine uterine tissue.
July 29, April 14, Based on internal test report RE, Data collected during 17 procedures for product introduction. April—June Postoperative morbidity and histopathologic characteristics of tonsillar tissue following coblation tonsillectomy in children: a prospective randomized single-blinded study.
Coll Antropol. Based on internal test report RE and RE, Independent surgeon and nurse feedback collected during Medtronic-sponsored labs. January and February Based on internal test report RE, Porcine testing. March 23, Based on internal test report RE collected during Medtronic-sponsored tissue testing conducted on March 23, 8.
Comparison of three common tonsillectomy techniques: a prospective randomized, double-blinded clinical study. Comparison of postoperative pain in pediatric patients undergoing coblation tonsillectomy versus cautery tonsillectomy. Otolaryngol Head Neck Surg. Based on internal test report RE, Tissue testing.
March 28, Jaws Cool Faster. February Fewer Instrument Exchanges. More Possibilities. Respir Care. Incidence, location and reasons for avoidable in-hospital cardiac arrest in a district general hospital.
In vivo assessment of an absorbable and nonabsorbable knotless barbed suture for laparoscopic single-layer enterotomy closure: a clinical and biomechanical comparison against nonbarbed suture. New laparoscopic suturing device: initial clinical experience. Stringer NH. J Am Assoc Gynecol Laparosc. Laparoscopic suturing evaluation among surgical residents. J Surg Res. Comparison and analysis of laparoscopic intracorporeal suturing devices: preliminary results.
J Endourol. Effect of a disposable automated suturing device on cost and operating room time in benign total laparoscopic hysterectomy procedures. Brown, S. Utilization of a porcine model to demonstrate the efficacy of an absorbable barbed suture for dermal closure. A multicenter randomized controlled trial comparing absorbable barbed sutures versus conventional absorbable sutures for dermal closure in open surgical procedures.
Aesthet Surg J. An in vivo comparison of barbed suture devices and conventional monofilament sutures for cosmetic skin closure: biomechanical wound strength and histology. Based on internal test report R, ASA verification testing with slow speed force limit evaluation.
January Based on internal test report , Final analysis of staple line vascularity using MicroCT. April 27, March November June Preclinical results may not correlate with performance in humans. Compared to uniform staple closure and pivoting anvil reloads. December Mansfield, MA: Medtronic; March 12, Significance of surgical plume obstruction during laparoscopy.
Boulder, CO: Medtronic; July Robotic versus laparoscopic stapling during robotic Roux-en-Y gastric bypass surgery: a case-matched analysis of costs and clinical outcomes.
Surg Endosc. Barbed versus conventional suture: A randomized trial for suturing the endometrioma bed after laparoscopic excision of ovarian endometrioma. J Minim Invasive Gynecol. A randomized study comparing traditional monofilament knotted sutures with barbed knotless sutures for donor leg wound closure in coronary artery bypass surgery. Interact Cardiovasc Thorac Surg. Aesth Surg J. July 18 to 26, August Based on internal test report RE rev A, Sabre blade comparison report.
Both devices are approved by the FDA to seal a 7mm vessel. TissueFect sensing technology to precisely manage energy delivery. BloomfieldCT Application of a Impact for an emergency splenectomy. Power remains constant over a wide range of tissue types.
FORCETRIAD COVIDIEN PDF
My product interest…